An automated comprehensive isotachophoresis−capillary zone electrophoresis ( ITP−CZE) system is described. The sample is focused in the first capillary by ITP . Instrumentation and fractionation of human serum proteins. An automated free- solution isotachophoresis system (FS-ITP) for preparative fractionation of. An automated free-solution isotachophoresis system (FS-ITP) for preparative fractionation of biopolymers is described, operated in a batch mode. The dimension.
|Published (Last):||16 October 2004|
|PDF File Size:||6.52 Mb|
|ePub File Size:||2.29 Mb|
|Price:||Free* [*Free Regsitration Required]|
This article has multiple issues. For reproduction of material from all other RSC journals and books: Multiple analytes with sufficiently different ionic mobilities will form multiple plateau zones.
If present in sufficient amounts, focusing analyte ions can displace all electrolyte coions, reaching isotachophorssis plateau concentration. For reproduction of material from NJC: XX is the XXth reference in the list of references. It alleviates the limitation of conventional ITP that it has limited separation capacity because of analyte zone overlap. The analytes of interest have intermediate ionic mobility.
There was a problem providing the content you requested
Lab on a Chip. This page was last edited on 5 Novemberat Using this device we demonstrate focusing and detection of a fluorescent species with a limit of detection of pM.
Back to tab navigation Fetching data from CrossRef. Jump to main content.
Miniaturized system for isotachophoresis assays – Lab on a Chip (RSC Publishing)
ITP is a displacement method: For reproduction of material from PPS: The article was received on 10 Maraccepted on 25 May and first published on 23 Jun Search articles by author G. If you are not the author of this article and you wish to reproduce material from it in a third party imstrumentation publication you must formally request isotachophoresi using Copyright Clearance Center. Reproduced material should isotachophordsis attributed as follows: We present an inexpensive hand-held device g that implements microchip isotachophoresis ITP with laser induced fluorescence LIF detection.
Back to tab navigation Download options Please wait Isotachophoresis ITP is a technique in analytical chemistry used for selective separation and concentration of ionic analytes.
Learn how and when to remove these template messages.
Information about reproducing material from RSC articles with different licences is available on our Izotachophoresis Requests page. November Learn how and when to remove this template message.
Authors contributing to RSC publications journal articles, books or book chapters do not need to formally request permission to reproduce material contained in this article provided that the correct acknowledgement is given with the reproduced material.
Back to tab navigation.
Isotachophoresis – Wikipedia
From this phenomenon ITP has obtained its name: This article relies largely or entirely on a single source. Jump to site search. If you are the author of this article you do not need to formally request permission to reproduce figures, diagrams etc.
Application of an electric potential results in a low electrical field in the leading electrolyte and a high electrical field in the terminating istoachophoresis.
This self-contained instrument integrates the functionality required instgumentation high voltage generation onto a microelectronic chip, includes LIF detection and is powered by a universal serial bus USB link connected to a laptop computer. The sample is introduced between a zone of fast leading electrolyte LE and a zone of slow terminating or: We find good agreement between simulations and experimental results. Retrieved from ” isotachophorrsis Details about the sub-components of the hand-held instrument PDF K.
In conventional ITP separations, a discontinuous buffer system is used.
We show that the response of the detector is linear with knstrumentation initial analyte concentration, making this device suitable for quantitative analysis. In all cases the Ref. Fetching data from CrossRef.
Please help to improve this article by introducing more precise citations. Previous Article Next Article. In peak mode ITP, analytes amounts are insufficient to reach plateau concentrations, instrumentztion analytes will focus in sharp Gaussian -like peaks. When all of the TE ions are dissolved, the focusing process ceases and the analytes are separated according to the principles of zone electrophoresis.
Please help improve it or discuss these issues on the talk page. Indeed, plateau mode ITP separations are readily recognized by stairlike profiles, each plateau of the stair representing an electrolyte or analyte zone having instruentation LE to TE increasing electric fields and decreasing conductivities.