Description: The DU Series Spectrophotometer is a microprocessor- controlled spectrophotometer intended for use in quantitative and qualitative biological. The Beckman DU Spectrophotometer is an efficient stand alone system that comes with complementary accessories to produce accurate results. Beckman DU UV/Vis Scanning Spectrophotometer. This refurbished DU Spectrophotometer unit; PDF Manual; Mouse; Power cord.
|Published (Last):||6 March 2011|
|PDF File Size:||4.22 Mb|
|ePub File Size:||19.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
Retrieved 30 August The DK-1 used a non-linear scroll, and the DK-2 used a linear scroll to automatically record the spectra.
Remove your samples from the area when you have finished using the machine. If you feel I did not meet your expectation in certain areas of the service, please kindly send me an email on how you feel and how i can improve manul area of service.
When they turned down the opportunity, National Technical Laboratories designed its own optical system, including both a control mmanual and a quartz prism. Infrared spectrophotometers were better suited than UV—Vis spectrophotometers to the analysis of C 4 hydrocarbonsparticularly manuao applications in petroleum refining and gasoline production.
Customer satisfaction is our 1 priority. The buyer is responsible for all shipping charges i. Freight – Check the item description or contact the seller for details. When you switch a lamp on, allow the machine to warm up for a half-hour before use.
Beckman series manuals here; | Hach Company/DU Questions and Answers on LabWrench
In developing the Model D, Beckman took the direct-coupled amplifier circuit from the pH meter and combined the optical and electronic components in a single housing, making it more economical.
In Morris, Peter J.
Brown states that it “revolutionized the measurement of light signals from samples”. The A, B, and C prototype models all coupled an external Beckman pH meter to the optical component to obtain readouts.
Then information from the plates had to be combined to create bcekman graph showing the spectrum as a whole. Spectroscopic methods for observing the absorption of electromagnetic radiation in the visible spectrum were known as early as the s.
Missouri City, Texas, United States. Retrieved 9 March Ships from United States. National Museum of American History. The DU enabled researchers to perform easier analysis of substances by quickly taking measurements at more than one wavelength to produce an absorption spectrum describing the complete substance. Technical Setup applied to both of the following applications 1. Seller does not offer returns. The slit mechanism was continuously adjustable from.
DU Date Transfer Procedure
One of the first fully integrated instruments : Retrieved 1 September The first prototype Beckman spectrophotometer, the Model A, was created at National Technologies Laboratories in Visit my eBay store. If you have to turn on the DU, the main power switch is located on the rear of the instrument. The item you’ve selected was not added spectrpohotometer your cart.
Confirm that these indicators are displaying: Chemical technology 1st English ed. The Model B’s rotary cell compartment was replaced with a linear sample chamber.
In the s, Beckman Instruments developed the DR and the DK, both of which were double-beam ultraviolet spectrophotometers. Learn more – opens in new window or tab. Return to the Main Menu when you are finished using the machine. Food and Drug Administration and state and local regulatory agencies.